ISALI 0-5mH2O 6m Cable Submersible Level Transmitter Level Transducer Level Sensor.
- 4. Every delicate chip are from the imported OEM parts, so as to keep the long haul stability.
0-5mH2O 6m Cable Submersible Level Transmitter Level Transducer Level Sensor
- 1. Keep from the flood voltage, over-voltage, over-current and the assurance of turned around extremity;
- 2. Do the test to ensure the capacity of over-burden opposition and stun resistance,precisely guarantee the over-burden limit;
- 3. Actualize the electromagnetic obstruction protection from stay away from the issue occurred under the condition of complicated electromagnetism condition;
- 4. Every touchy chip are from the imported OEM parts, so as to keep the long haul soundness.
- Estimation Range: 0-5mH2O/50kPa;
- Link Length: 6 meters
- Estimation Medium: Water;
- Power Supply: 24V DC;
- Yield Signal: 4-20mA;
- Precision: 0.5%FS;
- Over-burden Capacity: Two times full-scale weight;
- Utilizing Temperature: – 20℃-120℃;
- Remuneration Temperature Range: – 10℃-80℃;
- ZTC: (average): 0.02%FS/℃;
- STC: (average): 0.02%FS/℃;
- Long haul Stability: (typical):0.1%FS/year;
- Waterproof Class: IP68;
- Load Resistance: ≦(U-10)/0.02ω;
- Shell Material: 316L tempered steel;
- Blast evidence Type: Exia II CT6
- 1. The estimation medium ought to be good with hardened steel;
- 2. Try not to associate the wires when the power is on,connect the power in the wake of checking the association wires;
- 3. Shell and power ought to be associated with earth;don’t change the welded parts and associated wires of the outgoing items when client utilize it;
- 4. Sensor is identified with the exactness device,don’t disassemble,and what’s more,don’t contact the diaphragm,in case of break.
1 x 6m Cable Transducer Level Sensor